a) voltage： 380V±38V；
b) Number of phases: three-phase five-wire system (U, V, W phase line; neutral line; safety ground);
c）Power frequency: 50Hz ± 1Hz;
d) Power supply resistance: less than 0.5Ω.
Input power: ≤ 35kW
2、 Photography conditions
Tube voltage: 40 ~ 125 (step length 1kV)
Tube current: 16 ~ 200mA (12 files)
Current time product range: 0.5mAs ~ 320mAs
3. Tube voltage and tube current adjustment range (see Table 1)
4、 Working frequency：
Machine operating frequency: ≥ 40kHz.
5、 Current time product range：
0.5mAs ～ 320mAs
6、Overcurrent release characteristics: 380V, 40A.
7、 Radiography table
Table size: 1600 × 730 × 700 (mm) (length × width × height)
Table movement range: vertical 800mm horizontal 230mm
Table to film distance: 85mm
8、X-ray tube assembly, column
Horizontal movement (along the radiography table): 1620 mm
Move up and down: 1200 mm
Rotate around the center of the arm: ± 180o
9、Maximum size of film holder: 356mm × 432mm (14 “× 17”)
Stroke: 690mm Grid density: N40 Focal: 1000mm
1、 Compact structure design, excellent coverage performance, support full-body multi-angle radiography
2、 Easy to operate, accurate positioning, optimize the workflow.
3、 X-ray tube components up and down and along the table movement, radiography positioning more convenient.
4、No track and ground track, the integration of table, the installation is more simple.