For hospital wards and emergency room photography dedicated
Flexible mobile operating performance
Wireless remote exposure, greatly reducing the dose of physician radiation
二、The main technical parameters：
1 、Power supply conditions:
Power frequency： 50Hz±0.5Hz；
battery capacity： ≥5kVA；
Power supply allows maximum internal resistance：１?
2 、Photography conditions：
Tube voltage：40-90kV a total of 11 files, each file 5kV;
Tube current：30,50mA total 2 files
Time：0.05s-6 s a total of 20 files.
When the X-ray generator window down, the focus and the distance from the ground range of activities 450mm ~ 1770mm;
X-ray generator around the horizontal axis of ± 180o;
The X-ray generator has a rotation angle of ± 90o around the vertical axis;
4、Overall dimensions: 1290 (length) × 630 (width) × 1935 (high) (mm)