The combined adjusting step of image intensifier TV system
1. Do the image intensifier and the test table for cleaning
2. Put the image intensifier on the test table
3. To monitor, image intensifier and camera electrify
4, adjust the electronic circle between 195-197mm, the height of the gasket adjustment camera.
5. The sharpness of the image intensifier: the requirement is more than 14lp/cm, the actual 18lp/cm is reached, the test card puts the sharpness cut into 45 degrees, and the camera button is adjusted to the best image modulation
6, gray level: requirements greater than 8
7, adjust the fixed flange and camera plate, adjust the electronic circle to the middle of the display screen.
8, take out the test card, the inside of the electronic circle should be spotless. If there are stains, clear the output screen and camera lens.
9. Cover the camera cover, clean the camera cover and fix the camera lid.
The image intensifier TV system’s coordination work has been completed.
Author: X Ray Machine Accessories