Home›Blog ›X ray exposure hand switch design principle
X ray exposure hand switch design principle
Newheek X-ray exposure hand switch belongs to the field of medical equipment. The invention is composed of a control circuit and an exposure protection circuit, wherein the control circuit is connected with each control signal of the X-ray generator, and outputs an exposure preparation control signal and an exposure control signal; each control signal of the exposure protection circuit and the X-ray exposure hand switch The exposure preparation control signal and the exposure control signal outputted by the control circuit are connected, and the output exposure preparation protection signal and the exposure protection signal respectively control the exposure preparation circuit and the exposure circuit of the inverter control circuit in the X-ray exposure hand switch. The invention can realize the cutting function of the radiation source under the condition that the fault occurs inside the x ray exposure hand switch and the control signal of the exposure hand switch interface is abnormal.
Below, let me introduce the Newheek L04 x ray exposure hand switch specification.
Adopt Omron Inner Switch Mechanical Life of 50,000,000 Times
Super ABS Housing Electric Life of 300,000 Times
Operation Voltage: 125V
Current : 3A
Housing Material: white, ABS plastic
Hand Switch Cable Core: White, Black, Red, Green
Ambient Temperature: -20°C-70°C
Relative Humidity: ≤ 93%
Atmospheric Pressure: 50-106kPa
If you want to know more about Newheek x ray exposure hand switch, please contact our customer service.