200mA Medical Radiography x-ray Machine
Main Technology Parameters
1. Power Supply Requirements
Power Voltage: 380/220V±10%
Power Frequency: 50Hz±1Hz
Power Internal Resistance: 0.9Ω(380V) 0.3Ω(220V)
Power Capacity: 20kVA
2. Radiography Conditions
kV: 50~100kV Continuously Adjustment
mA: 50(small), 50, 100, 150, 200, 300mA Step Adjustment
Exposure Time: 0.1~6.3s Step Adjustment
3. X-Ray Tube
Small Focal Point:1.0×1.0mm2
Big Focal Point: 2.0×2.0mm2
4. Radiography Capacity
5. Radiography Table
Table Size: 2000×735×700(mm)(Length×Width×Height)
Table Surface Moving Range: Vertical 800mm, Horizontal 220mm
The Distance Between Table Surface and Film: 75mm
6. X-Ray Tube Subassembly & Stand Column
Horizontal Movement: (Along the Radiography Table) ≥1100mm
Vertical Movement: 610mm
Circle Arm Center: ±180°
7. Radiography Max. Cassette Size:
Grid Density: N28
Focal Distance: 1000mm
1. Compact physical design, superior coverage performance, and support general multi-angle radiography.
2. Simple operation, accurate location, and optimize working procedure.
3. Power voltage, x-ray tube voltage and tube current are showed respectively, which is convenient for observation.
4. KV, mA and s parameters are interlock protective sectionally.