First of all, we need to know why the C-arm is of the type. C-arm is divided into large C and small C depending on the use and appearance.
Today, I mainly explain the small C, also known as C-arm, small C-arm, small C, orthopedic C, C-arm X-ray machine, C-arm machine, megapixel X-arm, mobile C-arm.
The main purpose:
1. Orthopedics: whole bone, reduction, nailing
2. Surgery: taking foreign bodies, cardiac catheters, etc.
Newheek also has a small C, Mobile C -Arm X-Ray Machine, let me introduce it to you today.
Compact design, light and handy, easy to move and operate. The pulsed expo-
Sure mode produces superb image contrast. Powerful digital fluoroscopy system, for
Specialized minimally invasive procedures and routine surgical interventions.
High tension generator:
Output power: 5kW
Invertor: 100kHz, PWM
Power supply: single phase ac 220V, 7.5kVA, inner resistor ≤0.8Ω
Radiography : 40-120kV, 18-80mA, 1-160mAs
Fluoroscopy : 40-120 kV, 0.4-4 mA
Pulse exposure: 40-120kV, 18-80mA, 15-50pulses/s according to imagine
Sensor, max 255 pulses in single exposure.
Display: 320×240 pixel LCD
Mr. V of India is also very interested in our C-arm and feels that our C-arm is very cost effective.
If you want to know more about Newheek C-arm, please contact our customer service.
Company:Weifang Newheek Electronic Technology Co., Ltd.
ADD:No.3 Factory Building of No. 13426 Yuqing East Street, Weifang, Shandong, China