At present, many users have inquired whether the veterinary 50MA X-ray machine is easy to use? Because the user is a layman and often doesn’t know which device to choose, I’ll introduce the 50MA X-ray machine for veterinary use this year:
First of all, clarify which pet objects are being photographed? For small pets or medium and large pets, generally a 50MAX X-ray machine for veterinary use for small pets, if it is a 100MA X-ray machine for veterinary use for pets of about 50-60 kg (and for large pets, 200MA X-ray machine for veterinary use is required) machine.
1. Power condition of 50MAX optical machine for veterinary use:
Power supply voltage: AC 220V±22V;
Power frequency: 50Hz±0.5Hz;
Power capacity: ≥5kVA;
The maximum allowable internal resistance of the power supply: 1Ω
2. 50MAX photo-mechanical photography conditions for veterinary use: tube voltage: 40-90kV in 11 gears, 5kV each; tube current: 30, 50mA in 2 gears;
Time: 0.05s-6 s, a total of 20 gears.
3. Mechanical properties of 50MAX optical machine for veterinary use:
When the X-ray generator window is downward, the distance between the focal point and the ground is 450mm～1770mm;
The rotation angle of the X-ray generator around the horizontal axis is ±180º;
The rotation angle of the X-ray generator around the vertical axis is ±90º;
4. The overall dimensions of the veterinary 50MA X-ray machine: 1290 (length) × 680 (width) × 1935 (height) (mm)