The DR digital gastrointestinal machine consists of X-ray tube components, medical diagnostic X-ray high-frequency and high-voltage generators, microcomputer consoles, remote control stations, diagnostic observation beds, power cabinets, medical X-ray beam limiters, and medical X-ray image enhancement TV systems , Filter and flat digital DR detector. The DR digital gastrointestinal machine can not only take digital film, but also take digital fluoroscopy. One machine has multiple functions and meets the needs of hospitals for comprehensive X-ray diagnosis.
The working conditions of DR digital gastrointestinal machine are as follows:
(1) The power supply voltage is three-phase 380V, 50Hz, sine wave, the allowable range of power supply voltage value is 380V±38V, and the allowable range of frequency value is 50Hz±1Hz.
(2) The power supply capacity is not less than 60kVA.
(3) The power supply resistance should not be greater than 0.17Ω.
(4) Rated power: 100kV, 500mA, 0.1s, 50kW.